NJ ATM 1 - Particulate Emissions
- Heated out of stack filter (M5).
- "Condensible" particulates pass through the filter and condense in impingers. Can readily be combined with a CPM Method.
- Tester may adjust out-of-stack filter temperature to suit source specific conditions.
- Particulate Matter (PM) -- those substances retained in the sampling nozzle/probe and on a heated filter after the removal of uncombined water.
- A minimum sample catch of 50 mg is targeted to calculate the minimum sample times
- An absolute minimum of 3 mg catch is required;
- Any expected catch of less than 50 mg will require the use of EPA Alternative Method 005 to process the samples.
- During sampling a filter is maintained at a specified temperature (248oF + 25oF)
- Do not use silicone grease - it can be soluble in the probe rinse and can bias the PM value high.
- Run the train isokinetically.
- Stainless steel nozzles should be used with great care, port scrapings can bias results high.
- Acetone must be shipped as a hazardous material. Please call for IATA codes if you need them.
- Samples may be shipped at ambient temperatures.
- It may prove more economical to keep acetone rinses separate, ship other samples normally, and the acetone as hazardous.
- Be sure to include a chain of custody for each shipping container.
- This method is the state of New Jersey version of Method 5 - Promulgated in the mid 70's.
- The Front Half acetone rinse is to be baked after evaporation - prior to weighing.
- No, please contact us if you are interested!