NOTE: This method does not include all of the specifications (e.g., equipment and supplies) and procedures (e.g., sampling) essential to its performance. Some material is incorporated by reference from other methods in this part. Therefore, to obtain reliable results, persons using this method should have a thorough knowledge of at least the following additional test methods: Method 1, Method 6C, and Method 25A.
|Total Organic Compounds||N/A||< 2% of span|
This method is applicable for the determination of total gaseous organic concentration of vapors consisting primarily of alkanes. Other organic materials may be measured using the general procedure in this method, the appropriate calibration gas, and an analyzer set to the appropriate absorption band.
Adherence to the requirements of this method will enhance the quality of the data obtained from air pollutant sampling methods.
A gas sample is extracted from the source through a heated sample line, if necessary, and glass fiber filter to a nondispersive infrared analyzer (NDIR). Results are reported as volume concentration equivalents of the calibration gas or as carbon equivalents.
Same as Method 25A, Section 3.0.
This method may involve hazardous materials, operations, and equipment. This test method may not address all of the safety problems associated with its use. It is the responsibility of the user of this test method to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to performing this test method. The analyzer users manual should be consulted for specific precautions to be taken with regard to the analytical procedure.
This method is often applied in highly explosive areas. Caution and care should be exercised in choice of equipment and installation.
Same as Method 25A, Section 6.0, with the exception of the following:
A nondispersive infrared analyzer designed to measure alkane organics and capable of meeting or exceeding the specifications in this method.
Same as Method 25A, Section 7.1. No fuel gas is required for an NDIR.
Same as Method 25A, Section 8.0.
Same as Method 25A, Section 9.0.
Same as Method 25A, Section 10.0.
The sample collection and analysis are concurrent for this method (see Section 8.0).
Same as Method 25A, Section 12.0.
Same as Method 25A, Section 16.0.